ATTEN ADSCAL vs Rigol DSe were the two options left for considering . Manual, tracking, automatic cursor measurement function;. I have recently bought an ATTEN ADSCAL+ oscilloscope. During to instructions, I get the message: “Firmware update failed, error 0xFE. Buy Now Only: US$, Buy ATTEN ADSCAL Digital Oscilloscope MHz at the ADS Series User Manual Size: ( KB).
|Published (Last):||20 April 2011|
|PDF File Size:||16.98 Mb|
|ePub File Size:||15.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
II General Safety Requirement Channel waveform and it’s FFT waveform display on split screen. Positive pulse width larger than pulse width setting Positive pulse width equal to pulse width setting Negative pulse width less than pulse width setting.
Auto Measurement button Xds1102cal When you take automatic measurements, the oscilloscope does all the calculating for you. One picture is worth a thousand words; this is a now a detailed, picture album.
Transaction was fine and shipped quickly Review by reds Quality.
This Chapter shows you how to operate following tasks: To do track cursor measurement, follow these steps: Trig level at limit! These cursors demonstrate the physical meanings of these measurements.
Edge, Video, Pulse, Slope, and Alternative. Cursor measure covers Manual mode, Track mode and Auto mode.
Normal Use this mode when you want to see only valid triggered waveforms; Shipping and handing Item location: Page 81 Picture Table Auto measure function 2-Voltage measure menu: To adx1102cal a waveform Take the CH1 for example: This mode is good for observing repetitive waveforms. Picture Operate Instruction: Picture Auto mode Table Auto mode function menu: Page 98 Picture Table Utility System function menu 2: All pre-trigger data has been acquired and the oscilloscope is ready to accept a trigger.
Picture Table Waveform play back function menu 2: Set to Fine Picture 4. Don’t have an account? Continue to acquire data until the waveform record is full. Load Recall stored mask setting. Voltage probe compensation output and ground.
Page 6 General safety summary Avoid Exposed Circuitry. When a trigger ads1102cl set up properly, the oscilloscope converts unstable displays or blank screens into meaningful waveforms.
Atten ADSCAL Oscilloscope Product Reviews
This aspect has been sent to the manufacturer and they said they would work on ,anual. Page Picture Table Waveform play back function menu 2: Page 82 Table Auto measure function4-Delay measure menu: Option Setting Instruction Invert Turn on invert function. Picture All models: The Printer may override your selection for best fit. I used them back ‘in the day’ at IBM.
Comments to this Manuals Your Name. Press das1102cal MATH button to display the waveform math operations. Sound Open the key-press voice.
Atten ADS1000 Series Manual
FFT waveform and its Channel waveform can display on split screen at the same time. Repeat steps 7 and 8. Set up the channel bandwidth limit Take the CH1 for example; the tested signal is a pulse signal with the high frequency surge: The volume exquisite and it is convenient for carrying.
Ayten readout on the top right corner which unit is sec, show us the horizontal position.
ATTEN ADS1102CAL Digital Oscilloscope 100MHz
The shapes with manuall arrows beside them represent the front panel knobs. Same with the software. Test the Humorous content and distortion in the system Show the Noise in the DC Power supply Test the atteh and pulse response in the system Analyze vibration Picture Save waveforms to USB flash drive, follow these steps: The oscilloscope displays the values on the top of the right screen.
My main ‘scope has been the excellent Tek A.
Press the top second option button. In split display mode, the screen is divided into two parts and each part mwnual divided eight divides in vertical direction. Contact the site with comments or questions. Trigger coupling affects only the signal passed to the trigger system. When Select trigger condition. Table Auto measure function4-Delay measure menu: See Picture Picture