Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||14 November 2014|
|PDF File Size:||12.8 Mb|
|ePub File Size:||17.78 Mb|
|Price:||Free* [*Free Regsitration Required]|
Has anyone done anything on interfacing with the A since Matt C’s last post? IIRC, the sweep parameters were something like this: The forums emoticons took over for a bit there. Message 2 of 6. Matt – you’re right, the commands for mannual A are completely different from the – probably because of its age.
A RF IMPEDANCE ANL [Obsolete] | Keysight (formerly Agilent’s Electronic Measurement)
However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at. If not, what would creating a driver for it entail?
Thanks for your help guys. I have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen.
Agilent / HP 4191A RF Impedance Analyzer
When I gave it a try, though, it worked fine. I think the would probably take the old commands, but the ALSO has new commands.
It’s 84MB, so I hope you have a fast link. Message 6 of 6. In the code above, MHz is an arbitrary frequency.
Solved: GPIB Coding for the HPA Impedence Analyzer – Discussion Forums – National Instruments
Message 1 manuall 6. Should I be looking at doing it with LabView or something else? I have not worked with that instrument, but the Agilent web site has the operating and service manual here. Message 5 of 6. Accepted by topic author Matt C.
Most Active Software Boards: Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Sorry, that last line of code there is supposed to read: In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: I am able to communicate with the device through the GPIB interface, and am also able to retrieve information.
Message 3 of 6. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:. Guess you never know until you try!
Good luck with it I suspect that this difference might be causing my problems. Would you mind posting the code you used on the A?
I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different.
Message 4 of 6. You really need the manual.