Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||4 September 2009|
|PDF File Size:||1.15 Mb|
|ePub File Size:||10.73 Mb|
|Price:||Free* [*Free Regsitration Required]|
Message 4 of 6. I have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen.
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
Message 5 of 6. Should I be looking at doing it with LabView or something else? I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different.
If not, what would creating a driver for it entail?
However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at. Message 6 of 6. Most Active Software Boards: I think the would probably take manial old commands, but the ALSO has new commands.
Guess you never know until you try!
IIRC, the sweep parameters were something like this: In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: You really need the manual. Message 2 of 6. Would you mind posting the code you used on mxnual A?
Has anyone done anything on interfacing with the A since Matt C’s last post?
A RF IMPEDANCE ANL [Obsolete] | Keysight (formerly Agilent’s Electronic Measurement)
When I gave it a try, though, it worked fine. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:. The forums emoticons took over for a bit there. I am able to communicate with the device through the GPIB interface, and bp4191a also able to retrieve information.
4191A RF IMPEDANCE ANL
Matt – you’re right, the commands for the A are completely different from the – probably because of its mqnual.
I suspect that this difference might be causing my problems.
It’s 84MB, so I hope you have a fast link. In the code above, MHz is an arbitrary frequency. Message 3 of 6. Good luck with it H4p191a, that last line of code there is supposed to read: Message 1 of 6. Accepted by topic author Matt C.
Agilent / HP 4191A RF Impedance Analyzer
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Thanks for your help guys. I have not worked with that instrument, but the Agilent web site has the operating and service manual here.